Method1. This standard is issued under the fixed designation D ; the number 1 These test methods are under the jurisdiction of ASTM Committee D20 on. Measure the impact resistance of plastic film, coated paper and other materials utilizing the free falling dart method ASTM D ASTM D test for plastic film impact resistance and free falling drop dart cover the determination of the energy that causes plastic film to fail under specified.
|Published (Last):||24 June 2007|
|PDF File Size:||20.70 Mb|
|ePub File Size:||20.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Give our test engineers a call today for help configuring the best test machine and accessories according to your standard. Evaluating how your products and services meet and exceed quality, safety, sustainability and performance standards.
One missile weight is employed for each group and missile weight is varied in uniform increments from group to group. It has been noted that certain materials may stretch so far as to bottom out at the base of certain test instruments without actually rupturing. Send us a request Wilton UK: Have a question about conducting the test for this standard? Some films exhibit consistency axtm the initiation energy is the same as the total energy.
Through our network of over 43, people in 1, laboratories and offices in countries, Intertek provides quality and safety solutions to a wide range of industries around the world. The values stated in parentheses are for information only.
Active view current version of standard. Tests on materials that do not break, for any reason, are not considered to be valid.
Drop Dart ASTM D1709, ISO 7765
Link to Active This link will always awtm to the current Active version of the standard. By this technique, a uniform missile weight x1709 is employed during test and the missile weight is decreased or increased by the uniform increment after test of each specimen, depending upon the result fail or not fail observed for the specimen.
The method to analyze this test data is called the “Bruceton Staircase” method. Need help or have a question? Auditing Consulting Sourcing Training. We offer machines, apparatus, testers, tensile grips, test fixtures, clamps, holders, jigs, faces, jaws, extensometers, environmental and humidity chambers, furnaces, baths and custom solutions for satm applications. The mounting bracket is adjusted to the appropriate drop height, and the dart is inserted into the bracket.
This test method can be used for films whose impact resistances require masses of about 50 g or less to about 6 kg to fracture them. Assurance Testing Inspection Certification. Enabling you to identify and mitigate the intrinsic risk in your operations, supply chains and business processes. It is sometimes desirable to know the relationships among asrm results derived by different test methods.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. Testing Evaluating how your products and services meet and exceed quality, safety, sustainability and performance standards.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
ASTM D Plastic Film Impact Resistance | Free Falling Drop Dart Equipment
Validating the specifications, value and safety of your raw materials, products and assets. The alternative technique to the ASTM D test method provides for testing specimens in successive groups of ten. By this technique, the missile weight employed during the test is decreased or increased by uniform increments after the testing of each specimen, depending upon the result fail or not fail observed for the specimen.
If a test specimen fails, the drop weight is decreased by one unit. Also, the ISO method does not allow the alternative testing technique described in Section 11 of this test method. Send us a request Need help or have a question? A number of test samples are then impacted to determine an appropriate starting point for the weight of the dart. Standards can be obtained from appropriate standards authorities.
What is Third Party Certification and Accreditation for the food industry? Depending upon the expected impact strength of the test sample, either Method A or Method B is chosen. Note 1—Tests on materials that do not break, for any reason, are not considered to be valid.
This test uses a single dart configuration and a single drop height, while varying the weight of the dart.
Film samples that can be cut to mm x mm 9″ x 9″ specimens A minimum of 30 specimens are required for the test Data: ASTM D specifies a range of applicability from about 0.